Instructions for calibrating Monnit wireless temperature sensors.
Did you know that you can calibrate a wireless temperature sensor to make it more accurate for your application? The Monnit® wireless temperature sensor is accurate to +/- 1°C (1.8°F) and calibration to the sensor's working environment provides even greater accuracy. To calibrate your sensors, open your favorite browser and login to your iMonnit® account at www.imonnit.com.
It will take the sensor up to 2 heartbeats to fully calibrate. Do not attempt any further calibrations until at least 2 heartbeats have elapsed. The calibration function is a linear calibration so it is recommended that you calibrate the sensor in the environment that it will be measuring. (It is not advised to calibrate a temperature sensor to room temperature and then place the sensor in a cooler or freezer.)