The Monnit ALTA Wireless Ultrasonic Ranging Sensor provides distance readings from the sensor horn to the surface immediately perpendicular. The sensor uses an ultrasonic pulse and calculates the distance from the perpendicular surface by calculating the time it takes for the ultrasonic pulse to return to the sensor horn. Please consider the following recommendations and instructions for a successful deployment when installing and operating the Ultrasonic Ranging Sensor.
- Install the horn as far away from the surface being measured as possible (a liquid that splashes or condenses into the sensor’s horn will cause errors or inaccurate readings.
- Inspect the sensor horn for liquid or material and clear it out to avoid errors or inaccurate readings.
- Ensure the liquid has a stable surface when the readings occur; otherwise, readings will not be accurate, and errors will occur. We recommend filling liquid tanks from the bottom to avoid waves that might affect the readings.
- Remove obstructions that will affect the ultrasonic pulse beam on the sides of the container or those introduced while the sensor takes a measurement.
The Ultrasonic sensor has a maximum and minimum distance. You will observe a “Range Error” if the surface being measured is too far, too close, or there is some other sort of obstruction of the pulse or unstable surface detected.
Industrial Ultrasonic Sensor range
Minimum Measurement Distance: 20 cm (~3.9 in)
Maximum Measurement Distance: 750 cm (~157 in)
Sensor resolution: 1 cm (~0.39 in)
AA Enterprise Ultrasonic Sensor range
Minimum Measurement Distance: 10 cm (~7.8 in)
Maximum Measurement Distance: 400 cm (~295 in)
Sensor resolution: 1 cm (0.39 in)
You will see a reading indicating “Reading Unstable” if the pulse is not reliably returned to the sensor horn. It’s generally caused by a disturbed surface (like water in a tank filled from the top). The ultrasonic beam is less reliable when measuring the distance to unstable surfaces.
The sensor will be delivered calibrated and will provide accurate readings. You may want to apply a different calibration to the sensor using the Calibrate tab in iMonnit. However, when testing the sensor, be sure to apply the default calibration.
Testing the sensor
To test the sensor:
- Set a more frequent (10 minutes or so) Heartbeat.
- Reset the sensor to the default calibration (Default tab in the Calibrate tab in iMonnit; ensure the sensor checks in and accepts the default calibration).
- Inspect and clear the horn of any debris or liquid inside the horn.
- Place the sensor horn 90 centimeters (~3 ft) directly perpendicular to a flat stationary surface.
- Observe the sensor report data over the next several Heartbeats. You should expect to see consistent readings within the resolution presented in iMonnit. Clear the path and remove obstructions to the beam if you see errors. If errors or erratic readings persist, please contact email@example.com.